|Inner Chamber Size:||230mm X 360mm||Aotoclave Size:||440mm(W)*400mm(H)*620mm(D)|
|Capacity Of Water Tank:||Approx 2.5 L||Operation Temperature:||5-40 Degree|
|Net Weight:||42 KGS||Noise Level:||Less Than 70dB|
|Nominal Power:||1750va||Relative Humidity:||Max 80%, Non Condensing|
Class B Dental Used Steam Sterilizers
Thank you for choosing the our steam sterilizers.
Prior to operating this instrument, please read the operations manual carefully and follow all installation instructions.
If E88 appears on the screen when power on or appears on the report, please call your dealer or local service maintenance. Your steam sterilizer need a regular maintenance.
The sterilizer described in this manual is intended for the sterilization for medical, dental, beauty, Vet and Tattoo fields. It operates automatically with 134°C and 121°C sterilization temperatures. This sterilizer is in compliance with the European Directive 93/42/CEE and it has been produced in accordance with the EN 13060.
|Chamber||Φ 230mm x 360mm|
|Rated Voltage||AC 220-240V 50Hz (110-130V 60Hz)|
|Circuit breaker||F16A/400V (F20A/400V for 120V)|
|Capacity of the distilled water tank||Approx 2.5 L (Water at level Max)|
|Approx 0.5 L (Water at level Min)|
|Noise level||<70 dB|
|Relative Humidity||Max 80%, non condensing|
|Atmospheric pressure||76 k/Pa - 106 k/Pa|
|3||Instrument tray rack||1|
|4||Instrument tray handle||1|
|5||Door adjustment tool||1|
5.1.1 Open the door and remove all of the inner contents for unpacking.
5.1.2 Connect the power cord to an outlet of the appropriate voltage.
5.1.3 Power on
The switch is located underneath the control panel on the front side of the machine.
After switching on, the machine turns on the LCD and shows the door position,
water level, working program, date, time and etc.
|Daily||Clean the door seal|
|Clean the external surface|
|Weekly||Clean the distilled water tank|
|Clean the sterilization chamber|
|Every month(depending on the use)||Clean the filter inside the chamber and tank|
|Every 3/6 Months (depending on the use)||Replace the bacteriological filter|
|Every year||Replace the door seal|
7.1 Clean the distilled water tank every week with isopropyl alcohol or a medical disinfectant.
7.2 Clean the chamber weekly.
7.2.1 Remove all trays and the tray rack from the chamber.
7.2.2 Clean the chamber with a smooth cloth saturated with distilled water.
7.2.3 Apply the same procedure for the trays and rack.
7.3 Replacement of the bacteriological filter.
7.3.1 The bacteriological filter is in the back of the sterilizer.
7.3.2 Unscrew the filter by hand anti-clockwise.
7.3.3 Place the new bacteriological filter.
7.3.4 Screw the new filter by hand clockwise.
7.4 Clean the door seal.
Clean the door seal weekly, with a smooth cloth saturated with the distilled water.